Planned one-day closure - NCAR Mesa Lab - 10/10/15 more info>
- UCAR Home
- About Us
- For Staff
Calibrating measurements made during high-speed flight: An application of the new EOL laser air motion sensor
Presented by: Al Cooper- Earth Observing Laboratory, National Center for Atmospheric Research
Abstract: Measuring properties of the atmospher from a high-speed aircraft such as the NSF/NCAR Gulfstream GV presents a challenge because the flight speed affects most sensors. Measurements often involve complicated corrections as a result: To measure temperature, corrections must be made for dynamic heating caused by the motion of the aircraft, to measure airspeed, measurements of dynamic pressurem ambient pressure and temperature are needed; corrections are often made to the measured pressure that depend on the airspeed and/or orientation of the aircraft;ect. There are no standards or reliable calibration preferences for any of these measurements that apply in flight, so uncertainty analyses involve complicated and multi-dimensional examinations of these interactions and of how flight conditions might influence measurements from otherwise carefully calibrated sensors.
A reliable reference for any of these interlinked measurements can be of great value in reducing measurement uncertainty. A new EOL instrument, a Laser Air Motion Sensor (LAMS), now provides such a reference. This instrument measures the Doppler shift in laser light returned from ambient aerosols and provides a measurement of airspeed with less than 0.1m/s uncertainty. In this tak, I will discuss how various basic measurements from the LAMS to demonstrate how the presence of an absolute measurement of airspeed reduces measurement uncertainties not only in wind but also in pressure and temperature. Uncertainity and precision in pressure, for example have been reduced to <0.3 and 0.1 hPa, respecively, by use of measurements fro mthis new instrument. When combined with current high-precision measurements of altitude via GPS, this makes it feasible to map mesoscale pressure fields with high precision. Measurements of temperature have also been calibrated in flight wiht uncertainty of less that 0.3 deg C.
An added bonus of the LAMS is that, when combined with calibrated pressure measurements, it can also provide a measurement of the speed of sounds and so of temperature. This measurement of temperature does not depend on any other temperature sensor on the aircraft and should remain valid in cloud.
Tuesday January 7, 2014
3:30pm Refreshments Served at 3:15pm